- Model MW-2016 is an Automatic Self Balancing Power Meter 0-18 GHz. It measures RF Power with the help of temperature compensated Thermistor Mount from 10;uW to 10mW full scale with 1 % accuracy from 10 MHz to 40 GHz (depend upon type of Thermistor Mount).
- With calibration factor control switch, MW-2016 normalizes the Power Meter reading to compensate the efficiency of used Thermistor Mount.
- MW-2016 has provision for substitution measurements and Power Meter calibration. Output is provided for recorder or Digital Voltmeter readout.
|Instrument Type||Model MW-2016 , Digital Power Meter|
|Power Range||Four Ranges With Full Scale Readings of 10uW, 1 00u.W, 1mW and 10mW|
|Accuracy||± 1 % of fullscale on Milliwatt range And ± 2.5% on Micro Watt Range|
|Fine Zero||Automatic, operated by Front Panel Switch|
|Zero Carryover||Less than 0.5% of fullscale when zeroed on most Sensitive Range|
|Calibration||1 1 -position Switch normalizes Meter Factor Control Readings to account for Thermister Range|
|Recorder||Proportional to indicated Power with output 1 Volt corresponding to fullscale 1 kOhm output Impedance.|
|Bridge Ouput||(VRF and VComp), Direct Connection to Thermistor Bridges: Used in Calibration and Precision power measurement|
|POWER||230v ± 10% AC 50Hz, 10 Watts|
|Standard Accessories||(i) Mains Cable – 1 No.
(ii) Spare Fuse – 2 Nos.
(iii) Instruction Manual -1 No.
|Optional Accessories||(i) Connecting Cable for Thermistor Mount at AT-211 B is available on extra cost.
(ii) Temperature compensated Thermistor Mount Model: HP-8478B, 10MHz – 18 Ghz (Make Agilient or else) may be supplied.